Myron L Meters - Save 10% on Myron L meters, Myron L Ultrapens, Myron L Ultrameters, Myron L conductivity meters, Myron L monitors
|How do I clean the conductivity cell cup on the portable meters?|
|With everyday sampling, the cell cup may build up a residue or film on the cell walls that may cause the readings to become erratic. Use a 1:1 mixture of a common household cleaner such as Lime-A-Way, CLR, Tilex, etc and DI water. Pour into conductivity cell cup and scrub with a q-tip. Be sure to get around all the electrodes and the thermistor probe.
On the portable analog meters, use an acid brush to scrub the cell cup. Let it set for about 10 minutes. Rinse the cell cup thoroughly with tap water, then a final rinse with DI water.
For a quick and easy guide to cleaning your Ultrameter or Techpro, watch this video: How to Clean The Ultrameter and Techpro Cell Cup
If you have a portable analog meter, watch this video: How to Clean The Portable Analog Meter Cell Cup
Return To Top
|How do I replace the pH sensor in my portable meter?|
|For a quick and easy guide to replacing the pH sensor in your Ultrameter or Techpro, watch this video: How to Replace The pH Sensor in your Ultrameter or Techpro
If you have a portable analog meter, watch this video: How to Replace The pH Sensor in your Portable Analog Meter
Return To Top
We'd like to get your feedback. We are dedicated to exceptional service, so tell us what you think! Tell Us Now